SimPlate for HPC
Heterotrophic plate counts, simplified
Easy-to-count wells on a round plate that correlates to plate count agar. Positives fluoresce under UV light.
- The SimPlate for HPC test is U.S. EPA approved and included in Standard Methods for Examination of Water and Wastewater.
- Standard method for SimPlate.
- Internal comparisons to Standard Methods (9215B) pour-plate method show a correlation coefficient (r) of 0.97.
- Easier to read and use than the pour-plate method.
- No agar preparation.
- Most Probable Number (MPN) counting range minimizes need for dilutions.
- Instant media preparation.
- Less than 2 minutes of hands-on time.
- Reduced read time.
- Correlates with U.S. EPA-approved, pour-plate method1using plate count agar and 48-hour, 35°C incubation.2
- Simplified procedure minimizes chance of contamination.
- Less subjective interpretation than pour plate.
- Less equipment to purchase and maintain.
- Increased productivity saves labor costs.
- Minimal Quality Control (QC).
How SimPlate for HPC works
Multiple Enzyme Technology Medium—the SimPlate for HPC medium uses IDEXX Multiple Enzyme Technology (MET) to detect heterotrophic plate count (HPC) bacteria in water. The reagent contains multiple unique enzyme substrates, each targeting a different bacterial enzyme. The most common enzymes of waterborne bacteria are all targeted.
All of the enzyme substrates produce the same signal when metabolized. Therefore, a given waterborne bacteria will be able to metabolize one or more enzyme substrates and produce a signal. In the case of SimPlate for HPC, the signal is fluorescence under 365 nm UV light.
After the sample and medium are added to the plate, the wells fill automatically as the user swirls the plate. No pipetting is needed.
Once the wells are filled, the plate is angled and excess sample/medium mixture is poured into the absorbent pad. The SimPlate takes advantage of surface tension to hold the proper amount of sample/medium mixture in each of its 84 wells as excess is poured off.
SimPlates are incubated upside down to prevent condensation from collecting on the lid. Since each well is isolated from the others, bacteria from one well will cause only that one well to fluoresce.
The number of wells that fluoresce is converted to a Most Probable Number (MPN) using the table provided with the product. The high number of wells (84) results in a very high counting range and precise 95% confidence limits. The counting range is from <1 to 738 per plate (more than double a standard pour plate) with the multi-dose SimPlate for HPC.
How to use
Learn how to use SimPlate for HPC
Add 100 mL sterile diluent to media vessel.
Add 1 mL sample to SimPlate.
Add 9 mL media to SimPlate.
Swirl SimPlate and pour off excess.
Invert and incubate for 48 hours.
Read results under UV light.
Frequently asked questions
Yes. SimPlate for HPC is U.S. Environmental Protection Agency (EPA) approved for compliance monitoring of drinking water, effective November 29, 2002.
As per the 40CFR Part 141.72, surface water systems must provide disinfection at the point of entry whether the water is filtered or not. The distribution system water must have a detectable level of disinfectant or an HPC level less than or equal to 500/mL.
Yes, the base and lid are gamma irradiated. There are 25 plates per plastic sleeve. Any unused plates should remain in the bag and the bag should be closed. Do not leave any unused plates exposed to the environment.
Resources & Tools
Tests & Accessories
Search the Water Tests & Accessories tool
Product number: 98-05760-01
Product number: 98-05761-01
Product number: 98-12860-00
IDEXX Water has reference materials and approval documents to support the many products in our water portfolio. Find the document(s) you need by selecting the link below.
Search the Reference & Regulatory Documents tool
Water Customer Support
IDEXX Laboratories, Inc.
One IDEXX Drive
Westbrook, Maine 04092 USA